I would just like to pass on an experience I had with my Hanna Ph/Temp tester. I have had it for a little over a year. I store it in storage solution after each use. The readings started drifting and would not stablize when I was using it. I tried recalibrating the meter and it did the same thing. I contacted Hanna and they said replace the probe.
In a last ditch effort I decided to take a toothbrush with some dish washing liquid and gentley brush the probe. I rinsed it off , re-calibrated , and now the meter works as good as new!
Its worth a try. Maybe it will save you a few bucks.